Nanohunter II is Rigaku's next generation benchtop total reflection X-ray fluorescence (TXRF) spectrometer
that enables high-sensitivity ultra-trace elemental analysis, in liquids or on solid surfaces,
to the parts-per-billion (ppb) level. Total reflection X-ray fluorescence spectroscopy is a method
by which an incident beam of X-rays just grazes the sample, delivering low-background noise,
high-sensitivity measurement of ultra-trace elements.
Due to increasingly stringent environmental regulations, there is rising demand for a simpler method of conducting analyses down to the ppb level with regard to, for example, arsenic (As), selenium (Se) and cadmium (Cd) contained in factory waste liquids and effluent streams. Using the NANOHUNTER™ II spectrometer, analysis down to the ppb level becomes possible, even with a very small sample size, merely by adding a drop of liquid to the sample carrier, drying it, and then performing the measurement. Quantitative analyses using internal standard substances can also be easily performed.
The NANOHUNTER II TXRF analyzer combines a fully automatic optical axis adjustment system that provides stable high-sensitivity analysis in an easily handled benchtop form factor that allows quick and trouble-free operation. With a high-power 600 W X-ray source, a newly developed mirror (optic) and a large-area silicon drift detector (SDD), the NANOHUNTER II spectrometer enables high-intensity, high-sensitivity measurements. The lower limit of Cd detection is 2 ppb. For As and Se in a liquid, a detection lower limit of 0.8 ppb or less has been achieved.
One important advantage is the high efficiency with which Kα lines can be excited. Until now, xciting Kα lines in atomic no. 48 Cd has been very challenging, so practitioners relied on analysis sing L-lines, which are difficult to measure. The newly developed optic employed in this device can be used with a high-power excitation source of about 30 keV, which makes it possible to measure Kα lines with high signal-to-noise ratios and clearly defined peaks (when measuring Cd). ide adoption of this capability is expected in fields such as the screening and analysis of factory waste liquids and the analysis of beverages, as such as wine.
In the area of analysis of solid surfaces, there is demand (primarily in the fields of thin films and thick films) for analyses that penetrate slightly deeper than the surface. For this kinds of analyses, a methodology called grazing incidence X-ray fluorescence (GI-XRF) is employed whereby the elements beneath the surface are excited by varying the incident angle of the X-ray source. Because the NANOHUNTER II spectrometer has the necessary functionality to vary the angle of incidence, it is possible to perform depth profile surface analyses. The GI-XRF technique is applicable to nanoscale research.